In ptychography we solve the so called phase problem. Rather than a phase contrast image in which we record the intensity resulting from phase interference, ptychography provides phase interference directly. This results in an extremely sensitive and dose efficient imaging modality.
Experimental ADF and SSB ptychography images of graphene collected simultaneously showing how much stronger the ptychography signal is compared to the ADF signal.
It can also outperform conventional TEM at low doses.
In direct ptychography (both SSB and WDD) the relative phases of diffracted convergent beam electron diffraction (CBED) disks are solved via their mutual interference with the direct beam in probe position reciprocal space.
Above is schematic of the overlap of the diffracted disks arranged as they would be in the detector plane from the first ring of spots in graphene, hence the hexagonal symmetry. The central disk is the direct beam. The other disks oscillate as the probe is scanned.
If we take the Fourier transform with respect to probe position we are able to isolate the interference of the disks corresponding to each spatial frequencies with the direct beam.
In probe reciprocal space we obtain phase and amplitude as a function of both scattering angle and spatial frequency. If we display the phase as a function of scattering angle for a single spatial frequency we obtain for the frequencies aligned along the horizontal something like the following, where the red and blue regions are π out of phase.
The disk outlines are retained to illustrate the locations of the diffracted disks, with bold used for the disks corresponding to the positive and negative of the spatial frequency displayed which are perfectly out of phase.
It looks much the same for the other two independent pairs of spatial frequencies, except the selected disks have rotated around.
Each double disk overlap (where the diffracted disk and direct beam overlap) also contains the amplitude of the spatial frequency. Here we will just show the one diagram.
In the SSB method, the phase and amplitude of each spatial frequency is determined by integrating over its double disk overlap region in probe reciprocal space.
Once the phase and amplitude are determined for each spatial frequency, performing Fourier transforming back to real space interferes all the frequencies and produces a phase image.
“Efficient Phase Contrast Imaging in STEM Using a
Pixelated Detector. Part 1: Experimental Demonstration at Atomic
T. J. Pennycook, H. Yang, L. Jones, A. R. Lupini, M. F. Murfitt and P. D. Nellist, Ultramicroscopy 151, (2015), p. 160.